i am currently on my second device. A biotronik implanted over 2 years ago. This replaced my original Medtronic device which I received in 2010, when it reached the end of its battery life. When they did the swap, they 're used the original Medtronic leads. A week ago I received a jolt from my ICD while at work. it knocked me off of my feet. No warning. Not pleasant. I had never experienced my ICE firing previous to this. I assumed it fired for cause. Well a 911 call and an ambulance ride later, I'm in the ER. It fired again. No more pleasant than the first time. The Biotronik del arrives and connects to the device..... Finds fractures in 2 leads which caused the unnecessary shocks. He turned off the defibrillator lead and the bottom pacemaker lead. Kept me overnight to monitor me. Good me that I should be fine until they can replace the leads. They called today and scheduled the lead extraction and replacement in 3 weeks. Also told me that my insurance would not be billed. Medtronic was paying for it since it was their leads. This makes me suspicious. I sense that Medtronic is essentially admitting fault. The leads are just over 8 years old. Was told they should last 12-15 years. I know a while back that Medtronic had some leads recalled but that was prior to 2010 when mine were implanted. It just seems odd that they are so quick to pay. Anyone else have a similar experience with medtronic?
You know you're wired when...
Your signature looks like an EKG.
I'm 35 and got my pacemaker a little over a year ago. It definitely is not a burden to me. In fact, I have more energy (which my husband enjoys), can do more things with my kids and have weight because of having the energy.